Mooney Viscosity Meter / Tester , Mooney Viscosity Measurement
High Precision Labrotary Mooney Viscometer is used for testing
reclaimed rubber or gross rubber viscosity, scorch and
vulcanization index, etc. Computer control, data display, automatic
calibration, it is a precision labrotary test instrument of rubber
industries. Under the strong support of advanced PC machine
WindowsXP, window 7 or other windows software, coupled with the
excellentuser software and advanced hardware didn't prepare make
the whole measurement become very convenient, quick, accurate nd
GB/T1232, GB/T 1233, ISO667, ISO 289 ,ASTM D 1646, JIS K 6300 etc.
1. It used monolithic rotor control, which include: host,
temperature measurement, temperature control, data acquisition and
processing, sensors and electrical chains and other components.
These measurements, temperature control circuit consists of a
temperature control device, platinum resistance, heater
composition, capable of automatic tracking power and ambient
temperature changes, automatically correct PID parameters to
achieve fast and accurate temperature control purposes. Data
acquisition system and mechanical linkage to complete the rubber
vulcanization process of force torch signal automatic detection,
automatic real-time display of temperature and settings. After
curing, automatic processing, automatic calculation, print
vulcanization curve and process parameters. Show curing time,
curing power Ju, also has a variety of audible alert.
2. It controlled by computer, the computer setting the parameters
of the direct control of test parameters rheometer. Display
real-vulcanization curve and temperature curve, store test results,
different adjustable comparison of test results and in a different
1. Easy to use.
2. Temperature can be increased up to 185 within 7.5min, and
recovery time when beginning, and recovery time when beginning test
will be less 2min. And temperature fluctuation will be within 0.1.
3. All instruments adopted the way of PID to control temperature.
The smallest size and weight in all of the same type instrument in
4. The die was designed to international standard ISO.
5. With lower noise when working, ≤55dB.
6. It is following ISO 289 ,ASTM D 1646, JIS K 6300 .
7. Can variable Temperature unit “C, “ F Degree.
8.Use window XP program connect to machine & and can setting
Mooney or scorch time and show Dynamically Mooney value or scorch
time on the graph.
|Range of temperature control||From room temperature to 200°C|
|Temperature fluctuation in die||≤±0.1°C|
|Temperature display resolution||0.1°C|
|Rate of temperature increase||Room temperature ~ 100°C 6min|
|Measurement range of torque||0-200 MV|
|Mooney test accuracy||±0.2MV|
|Frequency of rotor||2±0.02r/min|
|Temperature PID Control||+/- 0.1°C|
|Upper & Lower||Separate plate temperature control|
|Power supply||220 V ,1 Phase, 50 Hz 680W|